How do I adjust the LCD contrast on my TSG95?
The LCD contrast is controlled by R1 on the main board inside the instrument. Turning the potentiometer clockwise increases contrast while turning it counter-clockwise decreases contrast. Please refer to the diagram below for locating R1 on the main board:
This FAQ Applies to:
No product series
FAQ ID 63486View all FAQs »