Is it possible to use the 80E01 and 80E06 electrical sampling modules with the 80A03?
Yes, however there are limitations. Some of them are:
The SMA adaptor (2.41/1.85mm t 2.92mm) will need to be used.
With the SMA adaptors on the input connector, the provided semi-rigid cable will not fit. A custom cable (not available from Tektronix) is required. Use a short high-quality SMA cable or fabricate your own semi-rigid cable.
The bandwidth of the 80A03 is limited to 12Ghz, nominal
This FAQ Applies to:
No product series
FAQ ID 59661View all FAQs »